Read online or download PDF • Page 44 / 67 • KROHNE IFC Converter EN User Manual • KROHNE Equipment. Micro Precision provides expert calibration services for KROHNE instruments including KROHNE IFC D -. Select from options for the IFC D which. KROHNE 08/ Subject to change without notice. Applicable to. Software-Versions. ○ IFC _. / D. Display version. No. and.
|Published (Last):||15 February 2008|
|PDF File Size:||19.63 Mb|
|ePub File Size:||6.87 Mb|
|Price:||Free* [*Free Regsitration Required]|
Do not cross or loop cables inside the terminal compartment.
Use separate cable entries see below for power supply, field current cables, signal lines, outputs and inputs. The flowmeter housings protecting the electronic equipment from dust and moisture must always be kept closed. The selected creepage distances and clearances have been sized in conformity with VDE and IEC for contamination category 2.
Supply circuits and output circuits are designed to meet the standards of overvoltage classes III and II, respectively. Signal cable to be permanently laid. Connect shields via stranded drain wires.
What’s missing? Tell us about it.
Underwater and underground installation possible. Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
Passive mode External power source required for operation kkrohne the outputs. All operating data have been factory set in accordance with your specifications. Power the unit, and the flowmeter will immediately start process flow measurement. Setting ranges The ranges are dependent on the meter size DN and the flow velocity v: P Pulse output overranged. If installed and operated in accordance with these operating instructions, it will rarely present any problems.
KROHNE IFC Converter Short EN : Manual
V ariable area flowmeters. Vo r tex flowmeters. Lev el measuring instruments. Switches, counters, displa ys and recorders. These concise instructions do not include the following: The operator is, howeverobligated to icc note of these sections in the.
Page 2 Contents 1 1. Page 3 1 Electrical connection: Page 8 Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2.
Page 10 3 Start-up 3. Page 11 Standard factory settings Function 1. Page 14 Function 1.
Page 15 Function 3. Don’t see a manual you are looking for? Tell us what’s missing.
Tell us about it. Or point us to the URL where the manual is located. Your manual failed to upload